Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Published (Last):||16 October 2013|
|PDF File Size:||14.87 Mb|
|ePub File Size:||5.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the probe has been just connected, the screen indicates the remaining time of warm up then the measurement screen appears, indicates the concentration and places kkimo on a bargraph.
Kimo MP200 Thermo-Anemo-Manometer
Press Ok for launching measurement. It is thus necessary to enter the real temperature of use to obtain more precise results.
To print the manual completely, please, download it. Air velocity with Pitot tube: Enter dataset time and interval of time between 2 measurements by selecting Period with access key. Click on Esc to return to previous screen. When the solenoid valve is enabled, it runs every minute. You can choose between a planned or a continuous dataset.
Wire probes with Smart-plus system Wire probes with Smart-plus system are automatically recognized when connected to the instrument. Select other dataset using arrow keys t and u. Once pressing Autoz key, klmo zero is readjusted. Select Start for dataset launching.
KIMO MP P USER MANUAL Pdf Download.
It’s a community-based project which helps to repair anything. For measurement launching, click on OK with the access key. Page of 16 Go. This manual also for: If Other is selected, you must enter a value. You can choose between a planned or a continuous dataset. You can access this function selecting COmax with the access key CO max The CO is measured on an adjustable m;200, the maximum value measured in this period is called CO max.
Kimo MP 200 H Thermo Anemometer Manometer Air Velocity Pressure Sensor
I – Technical Specifications actoolsupply. In regards of Quality insurance norms, we recommend that the instruments are checked once a year. Probe Hold Alarms Params Veloci. Select compensation It is possible to modify the kimk of the compensation in temperature.
Summary screen of selected dataset is displayed.
Enter text from picture: This menu allows to view probe Probe display information plugged to C2, Module, C1 or wireless connections. Battery When battery indicator flashes it is recommended to change the batteries: On enabled mode, the probe is connected, the measurement is carried out and the value is displayed.
Average – Rec Recording Veloci. Configuration If you use kmo probes, you must enter type into the Configuration sub-function. With Pressure function, you can access to following sub-functions – Hold – see Air velocity – Config.
On disabled mode, the kimmo is connected, the measurement is not carried out and the value is not displayed. V — General informations Maintenance KIMO performs calibration, adjustment and maintenance of all your instruments to guarantee a constant level of quality of your measurements.
This sub-function allows to compensate for any long-term drifts of the sensing element by a manual adjustment of the zero.
When CO peak is selected, the period is diplayed 30 seconds by default. Configuration mimo Params Parameters – Avg. Remove the front part at the back of the instrument. To modify the period, press Period with the access key.
This datasets can’t be set using Datalogger Software. When the countdown is finished, the CO max is displayed.